Melt Flow Rate Instrument MIP meets the standards of GB/T3682-2000,as well as ISO1133-97,ASTMD1238,JISK7210 and other standards involving main parameters.The melt flow indexer is suitable for measuring melt mass flow rate(MFR)and melt volume flow rate(MVR) of plastic materials such as polyethylene,polypropylene,polystyrene,ABS,polyamide,fiber resin,acrylic,POM,fluorine plastic,polycarbonate.It is widely used in plastic departments.
This kind of instrument can be used to test the melt flow rate(MFR) figures of all kinds of plastic and resin in the viscous flow state. It is not only suitable for the high melting temperature engineering plastics of polycarbonate, polyarylsulfone , fluorine plastic and nylon but also suitable for some other low temperature plastic products which are madeup of polyethylene(PE), polystyrene(PS), polypropylene(PP), ABS resin, polyoxymethylene(POM) and polycarbonate. So it is not only widely used in the plastic production, petroleum and chemical industry but also in the relevant institutions such as scientific research units and the community inspection departments.
The melt flow rate instrument is suitable for measuring the thermoplastic melt mass flow rate following the determination in the standard ”GB/T3682-2000 thermoplastic melt mass and volume flow rate measurement”. The standard is equivalent to the requirement in the ISO 1133:1997 and can meet the ASTM D1238 test.
Considering the beautiful appearance and convenient operation,the desktop structure of the melt flow rate instrument makes the equipment performance reliably and stably. Besides, the high-precision control instrument using fuzzy PID algorithm for temperature control has the feature of high sampling precision and high speed control.
Outlet diameter：Φ2.095±0.005mm outlet length：8.000±0.025mm Charging cylinder diameter：Φ9.550±0.025mm
Charging cylinder length：152± 0.1mm
Piston rod diameter：9.475±0.015mm piston rod length：6.350±0.100mm 2.Standard testing load(a total of eight full weight)
3.Temperature range：conventional room temperature——400℃（500℃optional） 4.Constant temperature accuracy：±0.5℃
5.minimum temperature displayed：0.1℃ 6.Cutting method: manual or automatic cutting 7.Temperature recovery time: less than 4 minutes
8.Printed：A mini printer is used to print outputautomatically 9.Supply voltage：AC220V±10% 50HZ